- Research article
- Open Access
Stem loop-mediated isothermal amplification test: comparative analysis with classical LAMP and PCR in detection of Entamoeba histolytica in Kenya
BMC Research Notes volume 10, Article number: 142 (2017)
Entamoeba histolytica, the causative agent for amoebiasis is a considerable burden to population in the developing countries where it accounts for over 50 million infections. The tools for detection of amoebiasis are inadequate and diagnosis relies on microscopy which means a significant percent of cases remain undiagnosed. Moreover, tests formats that can be rapidly applied in rural endemic areas are not available.
In this study, a loop-mediated isothermal test (LAMP) based on 18S small subunit ribosomal RNA gene was designed with extra reaction accelerating primers (stem primers) and compared with the published LAMP and PCR tests in detection of E. histolytica DNA in clinical samples.
The stem LAMP test indicated shorter time to results by an average 11 min and analytical sensitivity of 10−7 (~30 pg/ml) compared to the standard LAMP and PCR which showed sensitivities levels of 10−5 (~3 ng/ml) and 10−4 (~30 ng/ml) respectively using tenfold serial dilution of DNA. In the analysis of clinical specimens positive for Entamoeba spp. trophozoites and cysts using microscopy, the stem LAMP test detected E. histolytica DNA in 36/126, standard LAMP test 20/126 and PCR 17/126 cases respectively. There was 100% agreement in detection of the stem LAMP test product using fluorescence of SYTO-9 dye in real time machine, through addition of 1/10 dilution of SYBR® Green I and electrophoresis in 2% agarose gel stained with ethidium bromide.
The stem LAMP test developed in this study indicates potential towards detection of E. histolytica.
Amoebiasis caused by protozoan Entamoeba histolytica is an important human gastrointestinal infection responsible for over 50 million amoebic infection cases with over 100,000 deaths annually . It is a leading cause of death only surpassed by malaria and schistosomiasis  with most of these cases being reported in the developing countries [3–5]. In Africa, the burden of amoebiasis is high with an estimated E. histolytica infection median rate of 796 per 100,000 people . Studies conducted in Kenya indicated prevalence of 6–11% of E. histolytica/Entamoeba dispar in children at selected hospitals [4, 5, 7] and 11–32% among adults . Moreover, in more recent studies, the prevalence of E. histolytica by qPCR was recorded at 15% in Bungoma County, Western Kenya  while a much lower prevalence of 0.4% was reported among children with vertically transmitted HIV infection . This data suggests that amoebiasis is a heavy burden among Kenyan population which is also plagued by other diseases such as malaria, HIV-AIDS, tuberculosis and other non-communicable diseases.
The E. histolytica infection causes several intestinal and extra-intestinal conditions with dysentery and liver abscess being the most common . The algorithm to diagnose amoebiasis is often complex due to the unsatisfactory sensitivity and specificity of available tests. Microscopy is widely used but has low sensitivity and cannot differentiate E. histolytica from the morphologically similar non-pathogenic species E. dispar and amphizoic E. moshkovskii . Stool culture followed by isoenzyme analysis has been used to differentiate species but the methods are time consuming hence impractical for use in the routine diagnosis. Antibody detection tests have been developed and used widely but their downturn is low sensitivity in early disease and inability to distinguish active infection from previous exposure . The E. histolytica antigen detection in the stool using ELISA tests [14–16] has proved more sensitive than microscopy, however cross-reactivity with E. dispar limit their application [17, 18]. This far the PCR method has been the most sensitive method for discriminating between E. histolytica and E. dispar. Indeed, several PCR tests have been developed [19–21] but despite the reported advantage of PCR tests in diagnosis of E. histolytica, the method has limited use in routine diagnosis of amoebiasis in Kenya due to associated cost.
In the last decade, a rapid DNA amplification test called loop-mediated isothermal amplification (LAMP) of DNA was developed . The technique is a novel strategy for gene amplification which relies on DNA polymerase with strand displacement activities. The LAMP technique has recently been applied in detection of human diseases such as malaria [23, 24], human toxoplasmosis  and meningitis  and has been hypothesized to revolutionize field based molecular test [27, 28]. The LAMP is well suited for E. histolytica diagnosis in endemic areas because it does not require expensive equipment to achieve amplifications, sensitivity is equivalent to that of PCR and time to results is approximately 1 h. Moreover, the large amount of products formed offers the use of different visual detection formats that are applicable in rural endemic areas. Indeed, LAMP has recently been used successfully to detect other human stool pathogens such as Ascaris lumbricoides , Clostridium difficile  and hookworms . Previously, LAMP tests for E. histolytica have been developed based on small subunit rRNA gene  and HLY6 gene . In 18S rRNA gene, the nested PCR detected 0.1–1 parasite per reaction compared to LAMP test which detected 1 parasite  and 2 ng/µl compared to 15.8 ng/µl (~5 parasites per reaction) for LAMP test using DNA for HLY6 target respectively . LAMP uses many reaction components which is a major cost in the developing countries . However, several companies have come up with ready to use commercial isothermal master mixes slightly reducing the cost burden and need for protracted optimization procedures. These include Optigene, UK (http://www.optigene.co.uk/products-reagents/) and EIKEN Chemical Co Ltd, Japan (http://www.eiken.co.jp/en/). The advantages presented by LAMP method as a potential point of use test calls for more attention in improving this platform for use in endemic countries. On this context,  reported improved amplification speed and sensitivity of C. difficile, Listeria monocytogenes and HIV LAMP tests through addition of a second reaction accelerating primers called stem primers (target the stem section of the LAMP amplicon). In addition, stem primers have been used recently to improve the sensitivity of Trypanosoma brucei gambiense LAMP test by ~100-fold compared to the standard LAMP test . The advantage of stem primers is that they can be used in multiplex with loop primers  without affecting test reproducibility. In this work, we report an improved LAMP test for E. histolytica with inclusion of stem primers.
The reference DNA sample of E. histolytica HM-1: IMSS was kindly provided by Dr. Graham Clark, Department of Pathogen Molecular Biology, London School of Hygiene and Tropical Medicine, UK. The DNA to check the test specificity was prepared from E. dispar and Giardia lamblia using commercial DNA extraction kit (Qiagen, Essex, UK).
All samples were collected from children who presented to three participating outpatient clinics and those admitted to the paediatric ward of Mbagathi District hospital, Nairobi were examined for the presence of E. histolytica. In order to improve sensitivity of microscopy in detection of E. histolytica cysts, the technique of formal-ether concentration was applied .
The DNA was prepared from 126 samples scored as positive for Entamoeba (E. histolytica, E. dispar and E. moshkovskii complex) using microscopy. Genomic DNA was extracted using QiAmp® DNA stool Mini kit (Qiagen, Crawley, West Sussex, United Kingdom) as per the manufacturer’s instructions with slight modifications. Briefly, 200 μl of fecal suspension was washed five times with distilled water. To this suspension, 1.4 ml of ASL buffer was added and subjected to five times thawing (80 °C) and freezing (−80 °C) to rupture the rigid cysts. The genomic DNA was eluted in 50 μl of nuclease-free water and stored at −20 °C until use.
The PCR test targeting the small-subunit rRNA gene was used  with some modifications. Briefly a 25 µl test was done and consisted of 1× PCR buffer, 1.5 mM MgCl2, 2 mM dNTPs, 0.5 U of Taq polymerase and 10 pmol of forward primer (EntaF) and reverse primer (EhR). These primers generate a 166-bp PCR product and are specific for E. histolytica. The reference DNA template was 2 µl of DNA and 3–4 μl for clinical samples. The amplifications were done in a PCR system 9700 thermal cycler (Applied Biosystems, UK) under the following cycling conditions: An initial denaturation step at 94 °C for 3 min, followed by 35 cycles each consisting denaturation at 94 °C for 1 min, annealing at 58 °C for 1 min and extension at 72 °C for 1 min. The final extension was at 72 °C for 7 min. Reactions were done in duplicates and the resulting amplification products were separated by electrophoresis in 2.0% agarose gel in 1 × Tris–borate-EDTA at 100 V for 45 min and visualized under UV light after staining with ethidium bromide.
Design of LAMP primers
Four sets of primers each recognizing ten distinct sections of E. histolytica 18S small subunit ribosomal RNA (18S rRNA gene) (Genbank accession number X64142) and hemolysin (HLY6) gene (GenBank accession number Z29969.1) were designed using Primer Explorer version 3 software (http://primerexplorer.jp/lamp3.0.0/index.html). The targets were chosen due to the reported specificity and high number of copies (~200 copies) for18S rRNA gene  and HLY6 (400 copies/cell) . The software designed the following primers: forward and backward outer primers (F3 and B3) and forward and backward inner primers (FIP and BIP). The loop forward and backward primers (LF and LB) and stem forward and backwards primers (SF and SB) were manually designed following the respective published primer characteristics [22, 35]. The primers were blasted for target specificity using the basic local alignment search tool (http://0-www.ncbi.nlm.nih.gov.brum.beds.ac.uk/BLAST). The designed tests consisted of F3/B3, FIP/BIP, LF/LB and SF/SB primer combination.
The 18S and HLY6 LAMP primers were first analyzed for detection of the reference E. histolytica HM-1: IMSS using standard LAMP conditions. The tests specificity was checked with closely associated pathogen DNA extracted from morphologically similar but non-pathogenic E. dispar and G. lamblia. The primer set(s) that passed these criteria were then analyzed using a tenfold serial dilution of control DNA and using the standard LAMP test conditions . The most sensitive primer set for each target was selected for further analysis. The new tests were labeled stem 18S and Stem HLY6 LAMP tests respectively (Table 1) and the selected primer sets were used to optimize respective LAMP test using Taguchi method . Briefly, four reaction components determined to have the greatest effect on LAMP reaction namely inner primers, loop primers, stem primers and dNTPs had their concentrations varied at three levels. The inner primer concentration was varied from 30 to 60 pmol, loop primers from 10 to 30 pmol, stem primers from 10 to 40 pmol and dNTPs from 1 to 3 mM respectively. The concentrations of each reaction component were arranged in an orthogonal array  and used to determine the amount of amplification product formed . This was followed by regression analysis to determine the concentration optima for each selected reaction component . Other reaction components included 1× ThermoPol reaction buffer contained 20 mM Tris–HCl (pH8.8), 10 mM KCl, 10 mM (NH4)2SO4, 2 mM MgSO4 and 0.1% Triton X-100. The Bst 3.0 DNA polymerase (New England Biolabs, MA USA) was 0.5 µl, betaine at 0.8 M and SYTO-9 fluorescence dye at 2.0 µM (Molecular Probes, Oregon, USA). The template was 2 µl of DNA. The LAMP reaction were performed for 60 min at 62 °C using the real-time PCR machine and data acquired on FAM channel followed by reaction inactivation at 80 °C for 5 min. Once the optimized reaction conditions were determined the reactions were duplicated using a thermocycler and a water bath that maintained temperature at ~61–63 °C. The template for clinical samples was varied from 2 to 4 µl. For comparative purposes, the published LAMP test based on small subunit rRNA gene  and HLY6 LAMP test  were included.
Detection and confirmation of LAMP product
The LAMP product was detected through fluorescence of SYTO-9 dye in real time PCR machine, through electrophoresis in 2% agarose gel stained with ethidium bromide and after addition of 1 µl of 1/10 dilution of 10,000× stock SYBR® Green I (Sigma-Aldrich, St. Louis, MO, USA). To confirm that E. histolytica LAMP test amplified the predicted product, melt peaks were acquired using 1 °C steps, with a hold of 30 s, from 62 to 96 °C  post amplification and through digestion of the resulting LAMP product using restriction enzyme DdeI (New England BioLabs, MA, USA) and following manufacturers recommendations.
Analytical sensitivity LAMP test
The analytical sensitivity for the stem 18S LAMP test was carried out in duplicates using a tenfold serial dilution of ~300 µg/ml of reference DNA from E. histolytica DNA. To cover different published LAMP tests formats [32, 33, 35]. The following primer combinations were used: (i) Stem LAMP test with outer primers, (ii) Stem LAMP test without outer primers, (iii) Standard LAMP with loop primers and (iv) Standard LAMP test without loop primers (Table 2). These formats were compared with published LAMP test (without loop primers)  and PCR targeting 18S rRNA gene .
E. histolytica LAMP optimum reaction conditions
The Taguchi method determined the optimal concentrations for the four reaction components in stem 18S LAMP test as 35 pmol for FIP/BIP, 18 pmol for loop primers, 23 pmol for stem primers and 2 mM dNTPs. The stem HLY6 LAMP test showed the most efficient reaction at 40 pmol for FIP/BIP, loop primers at 20 pmol, stem primers at 15 pmol and 1.5 mM dNTPs. Concentrations for other reagents were as reported previously . The optimum temperature for stem LAMP test was determined at 62 °C and 50 min being the reaction cut-off point. Stem18S LAMP test indicated superior sensitivity to stem HLY6 LAMP test hence the latter was not progressed in the analysis of clinical samples.
E. histolytica LAMP product
The optimized E. histolytica stem 18S LAMP tests with and without outer primers indicated similar exponential real time amplification curves (Fig. 1a) with post amplification melting temperature (T m) of ~86 °C (Fig. 1b). The LAMP products showed the ladder like pattern on the agarose gel indicating the formation of stem-loop with inverted repeats (Fig. 2b). On addition of SYBR® Green I, the positive product turned green and the negative ones remained orange (Fig. 2c). The DdeI restriction enzyme digestion of stem 18S LAMP test product indicated the predicted amplicons of 143 and 103 bp.
Analytical sensitivity of LAMP and PCR tests
The stem 18S LAMP tests (with and without outer primers) indicated identical detection limit of 10−7 (30 pg/ml) (Fig. 1a; Table 2) while the standard LAMP test (with loop primers) and published LAMP test (without loop primers) indicated detection levels ranging from 30 to 300 pg/ml (Table 2). The standard LAMP test (without loop primers) showed low sensitivity and was not include in further analysis (Table 2). The stem 18S LAMP test sensitivity was not altered when the stem primers were used either in their forward or reverse orientation and/or when the template was increased from 2 to 4 µl. The PCR test based on the same target showed detection limit of 10−5 (3 ng/ml) (Fig. 1). The stem 18S LAMP test sensitivity was reproducible using thermocycler and water bath and no cross reactivity was recorded with non-target DNA. The optimized E. histolytica stem 18S LAMP test with and without outer primers F3 and B3 showed reduction in reaction time (cycle threshold = C T ) value of ~11 cycles (Table 2) compared to the standard LAMP test targeting the same gene.
Results for clinical samples
The stem 18S LAMP tests with and without outer primers detected 36 (28.6%) while the standard and published LAMP tests detected 26 (20.6%) and 21 (16.7%) of E. histolytica DNA from samples scored as Entamoeba spp. using microscopy respectively (Table 3). We recorded intermittent non-specific products with some replicates for stem LAMP test with outer primers, in which case the replicates were repeated. The conventional PCR classified 18 (14.3%) as E. histolytica. Other LAMP tests formats were not used in sample analysis since they indicated inferior analytical sensitivity.
In the present study we have designed a rapid and visual LAMP assay for detection of E. histolytica. The stem18S LAMP test is a modification of the standard LAMP test through inclusion of stem primers and indicate superior analytical sensitivity and shorter reaction time to results and translate to a higher detection of pathogen DNA in clinical samples compared to the standard LAMP format (Tables 2, 3). The recorded superior sensitivity can be attributed to the multiplexing of two reaction accelerating primers (loop and stem primers) in a single reaction as compared to the standard LAMP format with and/or without loop primers. The loop primers accelerate the reaction by priming the sequence loops between FIP/BIP primers  while the stem primers accelerate reaction by targeting the stem section of the sequence . It is therefore the use of two reaction accelerating primers that exponentially increase the amount of LAMP product, hence reduction in reaction time and increase in sensitivity. Surprisingly the omission of outer primers did not affect the stem 18S LAMP test sensitivity, although the ladder like bands on agarose gel were less bright compared with the format with the outer primers. This may indicate formation of less product in the latter format but did not translate to less sensitivity in terms of pathogen DNA detection. Indeed, the products of the two LAMP formats were confirmed to be identical through acquisition of post amplification melt curves (Fig. 1) and through digestion of the product with restriction enzyme. The primary role of the outer primers is to displace the newly synthesized strands into a single strand making it available for extension by either inner primer  and do not form part of the final LAMP product. It appears that the remaining primers may have some strand displacement activity, although not as efficient as the outer primers. The possibility of omitting the outer primers gives more flexibility for positioning of the remaining primers .
It is not clear as to why the LAMP test based on the HLY6 gene showed low sensitivity (10−2) and low detection of PCR positive samples despite the reported higher number of copies (~400 copies) . One possibility is that the reference DNA and the Kenyan samples may have mutation on the HLY6 gene or on the sequence section targeted by the published primers hence poor priming. Sequencing of the HLY6 gene from Kenyan isolates may answer this question in future. The lower sensitivity of the published LAMP format  compared to stem LAMP format is attributable to absence of loop primers. Indeed, our identical LAMP format based on the same gene showed similar lower detection levels with the published format (Table 2). On addition of loop primers, this LAMP format analytical sensitivity improves by tenfold and translate to detection of more positive clinical samples (Table 3). The use of loop primers to accelerate LAMP tests is recommended  and has been demonstrated to significantly improve LAMP tests sensitivity and detection of pathogen DNA in clinical samples [42, 43]. The sensitivity of E. histolytica LAMP test is further improved in this study through multiplexing loop primers with stem primers. This sequential addition of primers resulting in improvement of LAMP test sensitivity is unequivocal demonstration that the reaction accelerating primers are critical to any successful LAMP test. The resulting product was easily detected using SYBR® Green I dye allowing visual inspection of results. The SYBR® Green I is cheap but the need to open the tube to add the dye risk contamination with amplicon. Further the dye is non-specific and binds to any double stranded DNA including primer-dimers. To increase the confidence of using non-specific dyes, rigorous test optimization is necessary to reduce formation of spurious products. In addition, the use of more negative controls is recommended to the increase the confidence limit.
The stem LAMP test classified 36 (28.6%) of 126 DNA samples as E. histolytica. More encouraging results were that all PCR positive samples were also positive with stem 18S LAMP test, indicating that both tests were detecting the same thing. In this study, the detection rate of E. histolytica was at 14.4% using PCR and is equivalent to that reported earlier of 13.3% . All LAMP formats showed detection range of 15.9–28.6% which indicates LAMP method is superior to classical PCR and is a good improvement towards diagnosis of amoebiasis. Similar superior sensitivity of stem LAMP format to PCR has been recorded in diagnosis of sleeping sickness .
This is the first study in Kenya to report the detection of E. histolytica using LAMP method. It is possible that the prevalence of E. histolytica is even higher since a large portion of samples remained un-identified and/or that the microscopically observed cysts belong to the morphologically similar but non-pathogenic E. dispar and E. moshkovskii. No tests were done to check the presence of E. dispar and E. moshkovskii. The world prevalence of E. dispar is reported to be nine times that of E. histolytica . If that phenomenon holds in the prevalence of this species in Kenya, then a large portion of the remaining 90 (71.4%) DNA could be E. dispar. Having methods that can accurately differentiate Entamoeba spp. will help estimate their prevalence in Kenya and avoid unnecessary chemotherapy in patients with non-pathogenic species. It should be noted that in amoebiasis, the reason to treat is based on demonstration of trophozoites and/or cysts in the stool, as such LAMP test may not be relied upon to make a treatment decision. Since LAMP test is faster to perform, the technique could form part of diagnostic algorithms for amoebiasis where LAMP test is used to select cases for further confirmation with PCR.
In this study:
A new stem 18S LAMP test which is a modification of the standard LAMP test through inclusion of stem primers was developed.
The stem 18S LAMP test recorded superior sensitivity and shorter reaction time to results.
The detection rate of E. histolytica using the new test was higher than prevalence recorded earlier.
It is therefore recommended that this new stem 18S LAMP test be part of diagnostic algorithms for amoebiasis.
acquired immune deficiency syndrome
stool lysis buffer
- C T :
enzyme-linked immunosorbent assay
human immunodeficiency virus
hemolysin 6 gene
Kenya Medical Research Institute
loop-mediated isothermal amplification
neglected tropical disease
polymerase chain reaction
- Tm :
World Health Organization
Bercu TE, Petri WA, Behm JW. Amebic colitis: new insights into pathogenesis and treatment. Curr Gastroenterol Rep. 2007;9:429.
Markell EK, John DT, Krotoski WA. Lumen dwelling protozoa. In: Ozmat S, editor. Markell and Voge’s medical parasitology. 8th ed. Mexico: Saunders Company; 1999. p. 24–89.
Samie A, Obi LC, Bessong PO, Stroup S, Houpt E, Guerrant RL. Prevalence and species distribution of E. histolytica and E. dispar in the Venda region, Limpopo, South Africa. Am J Trop Med Hyg. 2006;75:565–71.
Gatei W, Wamae CN, Mbae C, Waruru A, Mulinge E, Waithera T, et al. Cryptosporidiosis: prevalence, genotype analysis, and symptoms associated with infections in children in Kenya. Am J Trop Med Hyg. 2006;75:78–82.
Mbae CK, Nokes DJ, Mulinge E, Nyambura J, Waruru A, Kariuki S. Intestinal parasitic infections in children presenting with diarrhoea in outpatient and inpatient settings in an informal settlement of Nairobi, Kenya. BMC Infect Dis. 2013;13:243.
World Health Organization. WHO estimates of the global burden of foodborne diseases: foodborne disease burden epidemiology reference group. Geneva: World Health Organization; 2015. p. 2007–15.
Saidi SM, Iijima Y, Sang WK, Mwangudza AK, Oundo JO, Taga K, et al. Epidemiological study on infectious diarrheal diseases in children in a coastal rural area of Kenya. Microbiol Immunol. 1997;41:773–8.
Kagira JM, Maina N, Njenga J, Karanja SM, Karori SM, Ngotho JM. Prevalence and types of coinfections in sleeping sickness patients in Kenya (2000/2009). J Trop Med. 2011. doi:10.1155/2011/248914.
Easton AV, Oliveira RG, O’Connell EM, Kepha S, Mwandawiro CS, Njenga SM, et al. Multi-parallel qPCR provides increased sensitivity and diagnostic breadth for gastrointestinal parasites of humans: field-based inferences on the impact of mass deworming. Parasites Vectors. 2016;9:38. doi:10.1186/s13071-016-1314-y.
Matey EJ, Tokoro M, Nagamoto T, Mizuno T, Saina MC, Xiuqiong B, et al. Lower prevalence of Entamoeba species in children with vertically transmitted HIV infection in Western Kenya. AIDS. 2016;30:803–5. doi:10.1097/QAD.0000000000001002.
Rao S, Solaymani-Mohammadi S, Petri WA Jr, Parker SK. Hepatic amebiasis: a reminder of the complications. Curr Opin Pediatr. 2009;21:145–9.
World Health Organization. World Health Organization/Pan American Health Organization/UNESCO report of a consultation of experts on amoebiasis. Wkly Epidemiol Rec. 1997;72:97–9.
Korpe PS, Stott BR, Nazib F, Kabir M, Haque R, Herbein JF, et al. Evaluation of a rapid point-of-care fecal antigen detection test for Entamoeba histolytica. Am J Trop Med Hyg. 2012;86(6):980–1.
Abd-Alla MD, Wahib AA, Ravdin JI. Comparison of antigen-capture ELISA to stool-culture methods for the detection of asymptomatic Entamoeba species infection in Kafer Daoud, Egypt. Am J Trop Med Hyg. 2000;62:579–82.
Evangelopoulos A, Legakis N, Vakalis N. Microscopy, PCR and ELISA applied to the epidemiology of amoebiasis in Greece. Parasitol Int. 2001;50:185–9.
Gonin P, Trudel L. Detection and differentiation of Entamoeba histolytica and Entamoeba dispar isolates in clinical samples by PCR and enzyme-linked immunosorbent assay. J Clin Microbiol. 2003;41:237–41.
Furrows SJ, Moody AH, Chiodini PL. Comparison of PCR and antigen detection methods for diagnosis of Entamoeba histolytica infection. J Clin Pathol. 2004;57:264–6.
Visser LG, Verweij JJ, Van Esbroeck M, Edeling WM, Clerinx J, Polderman AM. Diagnostic methods for differentiation of Entamoeba histolytica and Entamoeba dispar in carriers: performance and clinical implications in a non-endemic setting. Int J Med Microbiol. 2006;296:397–403.
Pinheiro SM, Carneiro RM, Aca IS, Irmao JI, Morais MA Jr, Coimbra MR, et al. Determination of the prevalence of Entamoeba histolytica and E. dispar in the Pernambuco state of northeastern Brazil by a polymerase chain reaction. Am J Trop Med Hyg. 2004;70:221–4.
Hamzah Z, Petmitr S, Mungthin M, Leelayoova S, Chavalitshewinkoon-Petmitr P. Differential detection of Entamoeba histolytica, Entamoeba dispar, and Entamoeba moshkovskii by a single-round PCR assay. J Clin Microbiol. 2006;44(9):3196–200.
Khairnar K, Parija SC. A novel nested multiplex polymerase chain reaction (PCR) assay for differential detection of Entamoeba histolytica, E. moshkovskii and E. dispar DNA in stool samples. BMC Microbiol. 2007;7:47.
Notomi T, Okayama H, Masubuchi H, Yonekawa T, Watanabe AN, Hase T. Loop-mediated isothermal amplification of DNA. Nucleic Acid Res. 2000;28:E63.
Chahar M, Mishra N, Anvikar A, Dixit R, Valecha N. Establishment and application of a novel isothermal amplification assay for rapid detection of chloroquine resistance (K76T) in Plasmodium falciparum. Sci Rep. 2017;7:41119. doi:10.1038/srep41119.
Tegegne B, Getie S, Lemma W, Mohon AN, Pillai DR. Performance of loop-mediated isothermal amplification (LAMP) for the diagnosis of malaria among malaria suspected pregnant women in Northwest Ethiopia. Malar J. 2017;16(1):34. doi:10.1186/s12936-017-1692-4.
Sun XM, Ji YS, Liu XY, Xiang M, He G, Xie L, Suo JX, Suo X. Improvement and evaluation of loop-mediated isothermal amplification for rapid detection of Toxoplasma gondii infection in human blood samples. PLoS ONE. 2017;12(1):e0169125. doi:10.1371/journal.pone.0169125.
Lee D, Kim EJ, Kilgore PE, Takahashi H, Ohnishi M, Tomono J, Miyamoto S, Omagari D, Kim DW, Seki M. A novel loop-mediated isothermal amplification assay for serogroup identification of Neisseria meningitidis in cerebrospinal fluid. Front Microbiol. 2016;6:1548. doi:10.3389/fmicb.2015.01548.
Notomi T, Mori Y, Tomita N, Kanda H. Loop-mediated isothermal amplification (LAMP): principle, features, and future prospects. J Microbiol. 2015;53:1–5. doi:10.1007/s12275-015-4656-9.
Lee P. DNA amplification in the field: move over PCR, here comes LAMP. Mol Ecol Resour. 2017;17(2):138–41. doi:10.1111/1755-0998.12548.
Shiraho EA, Eric AL, Mwangi IN, Maina GM, Kinuthia JM, Mutuku MW, Mugambi RM, Mwandi JM, Mkoji GM. Development of a loop mediated isothermal amplification for diagnosis of Ascaris lumbricoides in fecal samples. J Parasitol Res. 2016;2016:7376207.
Wei C, Wen-En L, Yang-Ming L, Shan L, Yi-Ming Z. Diagnostic accuracy of loop-mediated isothermal amplification in detection of Clostridium difficile in stool samples: a meta-analysis. Arch Med Sci. 2015;11(5):927–36. doi:10.5114/aoms.2015.54846.
Mugambi RM, Agola EL, Mwangi IN, Kinyua J, Shiraho EA, Mkoji GM. Development and evaluation of a loop mediated isothermal amplification (LAMP) technique for the detection of hookworm (Necator americanus) infection in fecal samples. Parasites Vectors. 2015;8:574. doi:10.1186/s13071-015-1183-9.
Liang S, Chan Y, Hsia K, Lee J, Kuo M, Hwa K, et al. Development of loop-mediated isothermal amplification assay for detection of Entamoeba histolytica. J Clin Microbiol. 2009;47:1892–5.
Rivera WL, Ong VA. Development of loop-mediated isothermal amplification for rapid detection of Entamoeba histolytica. Asian Pac J Trop Med. 2013;6(6):457–61.
Njiru ZK. Loop-mediated isothermal amplification technology: towards point of care diagnostics. PLoS Negl Trop Dis. 2012;6(6):e1572. doi:10.1371/journal.pntd.0001572.
Gandelman O, Jackson R, Kiddle G, Tisi L. Loop-mediated amplification accelerated by stem primers. Int J Mol Sci. 2011;12:9108–24.
Njiru ZK, Mbae CK, Mburugu GN. Loop-mediated isothermal amplification test for Trypanosoma gambiense group 1 with stem primers: a molecular xenomonitoring test for sleeping sickness. J Trop Med. 2017. doi:10.1155/2017/8630708.
Nagamine K, Hase T, Notomi T. Accelerated reaction by loop-mediated isothermal amplification using loop primers. Mol Cell Probes. 2002;16:223–9.
Cheesbrough M. District laboratory practice in tropical countries part 1. 2nd ed. Cambridge: Press Syndicate; 2005. p. 196–8.
Huber M, Koller B, Gitler C, Mirelman D, Revel M, Rozenblatt S, et al. E. histolytica ribosomal RNA genes are carried on palindromic circular DNA molecules. Mol Biochem Parasitol. 1989;32:285–96.
Cobb BD, Clarkson JM. A simple procedure for optimising the polymerase chain reaction (PCR) using modified Taguchi methods. Nucleic Acids Res. 1994;22:3801–5.
Monis PT, Giglio S, Saint CP. Comparison of SYTO9 and SYBR Green I for real-time polymerase chain reaction and investigation of the effect of dye concentration on amplification and DNA melting curve analysis. Anal Biochem. 2005;340:24–34.
Njiru ZK, Mikosza AS, Matovu E, Enyaru JC, Ouma JO, Kibona SN, et al. African trypanosomiasis: sensitive and rapid detection of the sub-genus Trypanozoon by loop-mediated isothermal amplification (LAMP) of parasite DNA. Int J Parasitol. 2008;38:589–99.
Njiru ZK, Yeboah-Manu D, Stinear TP, Fyfe JA. Rapid and sensitive detection of Mycobacterium ulcerans by use of a loop-mediated isothermal amplification test. J Clin Microbiol. 2012;50(5):1737–41. doi:10.1128/JCM.06460-11.
ZKN, FM, GNM conceived and designed the study protocol. FM, EM, ZKN performed data collection and analysis. ZKN did data presentation. FM drafted the manuscript. ZKN, CKM, JK, EM, GNM critically reviewed the manuscript. Planning, coordination and supervision of the project was done by ZKM, GNM, JK and CKM. All authors read and approved the final manuscript.
I would like to acknowledge financial support from National Commission for Science, Technology and Innovation, NACOSTI, Kenya, http://www.nacosti.go.ke. The provision of extra materials was through Meru University of Science and Technology, Kenya. The views expressed in this article do not reflect the views of our respective institutions. This article is published with permission from the Director, KEMRI.
The authors declare that they have no competing interests.
Availability of data and materials
The datasets supporting the conclusions of this article are included within the article and its additional files.
Ethics approval and consent to participate
The study was approved by the Kenya National Ethical Review Committee (SSC No. 2834). All parents and/or guardians of participating children were informed of the study objectives and voluntary written consent was sought and obtained before inclusion. A copy of the signed consent was filed and stored in password protected cabinets at KEMRI. The samples were analysed anonymously.
This work was supported by grant from the National Commission for Science, Technology and Innovation, NACOSTI, Kenya, http://www.nacosti.go.ke. The funders had no role in study design, data collection and analysis, decision to publish, or preparation of the manuscript.
About this article
Cite this article
Mwendwa, F., Mbae, C.K., Kinyua, J. et al. Stem loop-mediated isothermal amplification test: comparative analysis with classical LAMP and PCR in detection of Entamoeba histolytica in Kenya. BMC Res Notes 10, 142 (2017). https://0-doi-org.brum.beds.ac.uk/10.1186/s13104-017-2466-3
- Entamoeba histolytica
- Loop-mediated isothermal amplification, LAMP
- Stem-LAMP test